However, the BEM 500 can do much more: In addition to the flow rate, two temperature inputs are also available. The respective sensors are installed directly in the flowmeters. With the measured temperature in the supply and return lines, the measured volume is compensated according to a stored curve of the medium.
As with the BEM 300, the BEM 500 uses complex algorithms to calculate the "effective" flow rate, less any return flows that may occur (e.g. due to pulsation in the line). As with the BEM 300, this requires the use of a second sensor on the flowmeter to detect the direction of flow.
A Modbus output is available for the output of all raw and calculated data.
- For differential measurement with flowmeters in the supply and return lines.
- Volume compensation based on measured temperature.
- Detection of flow direction and correction of measured values of occurring return flows.
- Calculation of the mass of flow rate and consumption based on measured temperature and volume plus manual density.
- Input of 7 calibration points possible (linearisation).
- Relay, pulse, analogue and Modbus output.
- Technical data
Technical data BEM 500 Power supply 24 VDC (+- 20%) Environment Operating temperature -20 to +70 °C Storage temperature -20 to +80 °C Housing Dimensions 116 x 116 x 118 mm Protection class IP65 Material Plastic Weight 1.0 Kg Assembly Wall-mounted, control cabinet installation Display LCD, 4 lines Signal inputs OMG BEG 43D, BEG 44, BEG 45 (+BEV 13), BEG 553A/54A, BEG 06(A), BEG 47D/E OMP BEG 56A, BEG 45 (+BEV 13), BEG 47G OME BEG 60A, BEG61A, BEG 62A, BEG 47C OMH BEG 44, BEG 45 (+ BEV 13), BEG 53/54, BEG 06 Temperature sensor Pt100 Signal outputs Impulse output 2x (max. 250Hz) Relay output 1x (250 VAC / 30 VDC) Analogue output 2x (4 bis 20 mA/ 0 bis 10 V) Bus output Modbus RTU (RS 232/485), Slave Description Number of sensors 2 measurement devices (with 2 sensors each) Display Display of flow rate. Total, accum. Total Additional functions Flow direction detection and backflow compensation, temperature compensation, mass calculation with manual density (as per DIN 51757), linearization of the calibration curve
- Operating instructions